A Series Product
Knife Sharpner - Arthur Thomas Type, Automatic
Knife Sharpner - A.O Spencer Type, Automatic
Slide Staining Machines, Autoclaves
Vertical, Autospy Table,
Tissue Processors - improved, Automatic
Tissue Processors - latest, Automatic
Tissue Processors - Super Deluxe, Automatic Tissue Processors
- Electra Model ( Microprocessor Based )
Tissue Processors - Electra Model (Microprocessor Based)
Under the guidance of eminent pathologists our R & D Division
has developed yet another unique model of Tissue Processor having
complete Programmer logic Controller for complete cycle. Timing
& stage sequence, Wax bath temperature safety system etc. are
all controller/operated by feather touch key board with Digital
Display. The entire unit looks very attractive with aesthetic appeal.
The data is retained in case of power failure. Front panel comprises
of the key board, digital display, control switch. The PID temperature
controller have been used for controlling the wax bath temperature.
The 12 stage timing sequence may be programmed for each stage for
duration between 1 minute to 9 hrs. 59 minute, in steps of 1 minute.
Delay time can be programmed upto 99 hours and will be sequenced
with main control.
Wax Bath Safety Control:-
Wax Bath PID temperature controller gives signal if temperature
is not between 50º to 65ºC. Safety System will function
at 10th, 11th & 12th position and will stop the main unit in
stage 9 onward till the set temperature is attained in the wax bath.
Slowly rotating 2 basket rotors are also operated through the microprocessor.
The unit is supplied complete with 2 Nos. S.S. Basket Rotors 2 Nos.
S.S. Baskets, 24 Nos. S.S.Tissue Capsules, 10 Nos. S.S. beakers
without lip, 2 Nos. S.S. Wax BAths (Special designed for use with
PID temperature controller)
Suitable to work on 220 V, single phase, 50 Hz, AC supply
Capacity:- 1 litre, 2 liters
Automatic Wax Bath specially designed for PId control and PT 100
sensor. It signal's the machine in stage 9, prior to immersion in
Wax and if wax is melted it automatically takes change from stage
9 to 10 and so on, but if wax is not melted due to some reasons,
then machine signals "Wax not melted or temperature is not
proper". When wax is melted/wax temperature is attained then
only the tissue basket is allowed to immerse or rotate in the wax
bath. Thus 100% safety of precious tissue.
Basket Rotor which rotates the tissue basket slowly.
Tissue capsule basket S.S. even perforation.
Tissue capsule S.S. 28 x 28 x 9 mm with sliding cover.
Tissue capsule S.S.20 x 20 x 10 mm with sliding cover.
Tissue capsule S.S. Circular, 36 mm dia. With auto presslid.
Divider for Tissue capsule 's' shape.
Divider for Tissue capsule 'v' shape.Beaker without lip.